Adjusts the external calibration constants for the analog output section of a DSA device.
You must use the device to generate a high voltage and low voltage at a specified gain, measure the high and low voltages, then specify the requested high voltage, low voltage, and gain along with the actual high voltage and low voltage.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
channel is the number of the channel to calibrate. This number is the numeric portion of the physical channel name, not the full physical channel name.
requested low voltage is the low voltage you attempted to generate at the gain setting you specified.
actual low voltage is the actual low voltage an external sensor measures.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
requested high voltage is the high voltage you attempted to generate at the gain setting you specified.
actual high voltage is the actual high voltage an external sensor measures.
gain setting is the gain setting you used when you attempted to generate the requested high voltage and requested low voltage.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application