This file contains important information about the NI-845x software for Windows, including installation instructions, new features, a partial list of bugs fixed, and known issues.
NI-845x software for Windows is required to use the NI USB-8451 and NI USB-8452 I2C and SPI interfaces.
The NI-845x Software and Hardware Installation Guide included with the NI-845x CD includes instructions for installing National Instruments hardware and software.
The primary document for NI 845x products is the NI-845x Hardware and Software Manual. This manual includes an introduction to I2C and SPI buses, specifications for hardware, information about how to get started with application development, and a complete reference for all NI-845x functions.
To view the manual, select National Instruments»NI-845x»NI-845x Documentation»NI-845x Hardware and Software Manual.pdf from the Windows Start menu or NI Launcher.
Links to the NI-845x Hardware and Software Manual are in the online help for each NI-845x function within the National Instruments LabVIEW and LabWindows™/CVI™ development environments.
NI-845x 17.0 supports the following operating systems:
The NI-845x installer does not support Windows 2000/NT/Me/98/95 or the Windows Server non-R2 editions.
Note In 2016 NI-845x dropped support for Windows Vista, Windows XP, Windows Server 2003, and installations of Windows 7 without any service packs. NI-845x 17.0 will not install or run on an unsupported OS. You cannot deploy or distribute applications that use NI-845x 17.0 to an unsupported OS. Additionally, after installing NI-845x 17.0, you cannot use any installers built on this computer with any version of LabVIEW, LabWindows™/CVI™, NI TestStand™, or Measurement Studio on an unsupported OS.
For more information about the changes to our OS support for 2016, refer to KB 79UC78LS, Why Does my LabVIEW, LabWindows/CVI, Measurement Studio, or TestStand Built Installer Fail on Windows XP/Vista and Server 2003?.
Note Support for Windows 32-bit operating systems may require disabling physical address extension (PAE). To learn how this might affect your system and what actions you might need to take, visit ni.com/info and enter the Info Code PAESupport.
1 NI software installs VC2015 Runtime and .NET 4.6.2. Windows 8.1 and Windows Server 2012 R2 require Microsoft updates to support these items. Refer to Microsoft KB2919442 and KB2919355 for more information about how to install these updates.
2 NI software is signed with a SHA-256 certificate. Windows 7 SP1, Windows Embedded Standard 7 SP1, and Windows Server 2008 R2 SP1 require Microsoft updates to support SHA-256. Refer to Microsoft KB3033929 for more information about how to install this security update.
This version of NI-845x supports the following application software versions, including service packs. If you are not using NI application software, refer to Microsoft Visual Studio Support.
|Application Software||Versions Supported by NI-845x|
|LabVIEW||2014, 2015, 2016, and 2017|
|LabWindows™/CVI™||2012 and higher|
NI-845x supports the following Microsoft Visual Studio versions and associated programming languages.
|Programming Languages||Visual Studio Versions Supported by NI-845x|
|Microsoft Visual C++||6.0 (or newer)|
Before installing the NI-845x software, log on as Administrator or as a user with Administrator privileges. The NI-845x software setup program must have Administrator privileges because the program modifies the configuration registry of your system. Complete the following steps to install the NI-845x software.
The following changes were made to version 17.0 of the NI-845x software compared to NI-845x 16.0:
The following changes were made to version 16.0 of the NI-845x software compared to NI-845x 15.0:
The following changes were made to version 15.0 of the NI-845x software compared to NI-845x 14.0:
The following changes were made to version 14.0 of the NI-845x software compared to NI-845x 2.1.2:
The following changes were made to version 2.1.2 of the NI-845x software compared to NI-845x 2.1.1:
The following changes were made to version 2.1.1 of the NI-845x software compared to NI-845x 2.1:
The following changes were made to version 2.1 of the NI-845x software compared to NI-845x 2.0.1:
The following changes were made to version 2.0.1 of the NI-845x software compared to NI-845x 2.0:
The following changes were made to version 2.0 of the NI-845x software compared to NI-845x 1.1.5:
The following changes were made to version 1.1.5 of the NI-845x software compared to NI-845x 1.1.4:
The following changes were made to version 1.1.4 of the NI-845x software compared to NI-845x 1.1.3:
The following changes were made to version 1.1.3 of the NI-845x software compared to NI-845x 1.1.2:
The following changes were made to version 1.1.2 of the NI-845x software compared to NI-845x 1.1.1:
The following changes were made to version 1.1.1 of the NI-845x software compared to NI-845x 1.1:
The following list details the hardware models supported in this version of NI-845x.
NI-845x software supports only National Instruments I2C and SPI hardware products. For information about software for I2C and SPI products from other vendors, refer to the documentation you received with those products.
The following items are issues known to be present in NI-845x. This is not an exhaustive list of known issues in the current version of NI-845x.
On Windows 8/7/Vista, access to the NI USB-845x I2C/SPI Interface is limited to a single process. This means you cannot control a single NI USB-845x I2C/SPI interface from multiple executables at the same time.
Clarification of the DRDY (data ready) SPI signal is discussed in a white paper on ni.com: SPI Stream Configuration using the TI ADS833x Example in the "Configuring CONV, DRDY, and CS" section. Visit ni.com/info and enter the info code TIADS833X to view this white paper.
The following items are the IDs and titles of a subset of issues fixed in a given release of the NI-845x software. If you have a CAR ID, you can search this list to validate that the issue has been fixed. This is not an exhaustive list of issues fixed.
The following items were fixed between version 2.1.1 and version 2.1.2 of the NI-845x software.
|Bug ID||Fixed Issue|
|390379||ni845xSpiScriptNumBitsPerSample is not exported from ni845x.dll.|
|402639||Deadlock when using DIO and SPI Read/Write in parallel loops.|
The following items were fixed between version 2.1 and version 2.1.1 of the NI-845x software.
|Bug ID||Fixed Issue|
|369991 and 371608||Running NI-845x on Windows XP with DAQmx 9.4 or older may cause LabVIEW to crash at startup.|
|374074||NI-845x System API Provider may crash if no NI-845x hardware is installed on the system.|
|385948||You may get error –301706 (kNi845xErrorInvalidDeviceId) the first time after running an NI-845x application with a USB-8451 or USB-8452 that has older firmware. After receiving this error, your hardware may show up with a different name.|
The following items were fixed between version 2.0.1 and version 2.1 of the NI-845x software.
|Bug ID||Fixed Issue|
|363149||Running an I2C script with an uninitialized clock rate on the NI USB-8452 no longer defaults to a 1 MHz clock. It now defaults to a 100 kHz clock.|
|358686||Streaming API: Tightened the timing between drive high/low signals and starting of the waveform to minimize the response to the first DRDY.|
|329125||The TI ADS833x EVM example has been rewritten to allow for selecting multiple channels.|
|245664||Fixed an issue that could cause the NI USB-8451 to hang when connected to a noisy SDA signal.|
The following items were fixed between version 1.1.5 and version 2.0 of the NI-845x software.
|Bug ID||Fixed Issue|
|278688||Refnum constants/controls cannot be created for configurations, making it harder to create subVIs.|
|207029||NI 845x runs at half speed with two devices.|
|898939||NI 845x does not reference count LabVIEW open calls.|
The following items were fixed between version 1.1.1 and version 1.1.2 of the NI-845x software.
|Bug ID||Fixed Issue|
|110899||Memory leak in NI-845x Create Config Reference and NI-845x Close Reference.|
For more information about NI-845x software, refer to the NI-845x Hardware and Software Help. To view the help, select National Instruments»NI-845x»NI-845x Documentation»845xAPI.chm from the Windows Start menu or NI Launcher.
You can automate the installation of most NI products using command-line arguments to suppress some or all of the installer user interface and dialog boxes.
If the NI product you are installing uses Microsoft .NET 4.0, the .NET installer may run before any NI software installs and may require a reboot before the installation of NI software begins. To avoid a .NET reboot, install .NET 4.0 separately before you install NI software.
For more information about automating the installation of NI products, refer to the following KnowledgeBase articles:
Microsoft Windows 10 is the latest version of the Windows operating system and features significant changes compared to previous versions. Windows 10 introduces several new capabilities and also combines features from both Windows 7 and Windows 8. For more information about NI support for Windows 10, visit ni.com/windows10.
When you install NI software on Microsoft Windows 8.1, you will notice a few additional tiles in the Apps view, including shortcuts to NI application software products such as NI LabVIEW, Measurement & Automation Explorer (NI MAX), and NI Launcher. For more information about NI support for Windows 8.1, visit ni.com/windows8.
© 2005–2017 National Instruments. All rights reserved.
Under the copyright laws, this publication may not be reproduced or transmitted in any form, electronic or mechanical, including photocopying, recording, storing in an information retrieval system, or translating, in whole or in part, without the prior written consent of National Instruments Corporation.
NI respects the intellectual property of others, and we ask our users to do the same. NI software is protected by copyright and other intellectual property laws. Where NI software may be used to reproduce software or other materials belonging to others, you may use NI software only to reproduce materials that you may reproduce in accordance with the terms of any applicable license or other legal restriction.
End-User License Agreements and Third-Party Legal Notices
You can find end-user license agreements (EULAs) and third-party legal notices in the following locations after installation:
U.S. Government Restricted Rights
If you are an agency, department, or other entity of the United States Government ("Government"), the use, duplication, reproduction, release, modification, disclosure or transfer of the technical data included in this manual is governed by the Restricted Rights provisions under Federal Acquisition Regulation 52.227-14 for civilian agencies and Defense Federal Acquisition Regulation Supplement Section 252.227-7014 and 252.227-7015 for military agencies.
IVI Foundation Copyright Notice
Content from the IVI specifications reproduced with permission from the IVI Foundation.
The IVI Foundation and its member companies make no warranty of any kind with regard to this material, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose. The IVI Foundation and its member companies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material.
Refer to the NI Trademarks and Logo Guidelines at ni.com/trademarks for information on NI trademarks. Other product and company names mentioned herein are trademarks or trade names of their respective companies.
For patents covering the NI products/technology, refer to the appropriate location: Help»Patents in your software, the patents.txt file on your media, or the NI Patent Notice at ni.com/patents.