Configures common properties for edge triggering.
When you initiate an acquisition, the digitizer waits for the start trigger, which is configured through the Trigger Source property. The default is immediate. Upon receiving the start trigger, the digitizer begins sampling pre-trigger points. After the digitizer finished sampling pre-trigger points, the digitizer waits for a reference trigger. Upon receiving the reference trigger, the digitizer finishes the acquisition after completing post-trigger sampling.
The method used to couple the trigger signal.
|AC||0||Couples the trigger signal with an AC current.|
|DC||1||Couples the trigger signal with a DC current.|
|HF Reject||3||Couples the trigger signal with a high frequency reject filter.|
|LF Reject||4||Couples the trigger signal with a low frequency reject filter.|
|AC Plus HF Reject||1001||Couples the trigger signal with an AC current and a high frequency reject filter.|
The trigger edge on which to trigger the digitizer.
Handle that identifies the NI-SCOPE instrument session as previously allocated by Initialize With Options.
Error conditions that occur before this node runs.
The node responds to this input according to standard error behavior.
Standard Error Behavior
|error in does not contain an error||error in contains an error|
|If no error occurred before the node runs, the node begins execution normally.
Default: No error
The source you want the digitizer to monitor for a trigger.
|Channel 0||Monitors Channel 0 for a trigger signal.|
|Channel 1||Monitors Channel 1 for a trigger signal.|
|Channel 2||Monitors Channel 2 for a trigger signal.|
|Channel 3||Monitors Channel 3 for a trigger signal.|
|Channel 4||Monitors Channel 4 for a trigger signal.|
|Channel 5||Monitors Channel 5 for a trigger signal.|
|Channel 6||Monitors Channel 6 for a trigger signal.|
|Channel 7||Monitors Channel 7 for a trigger signal.|
|External TRIG||Monitors an external trigger input for a trigger signal.|
Default: Channel 0
The voltage threshold for the trigger.
The length of time the digitizer waits after detecting a trigger before enabling NI-SCOPE to detect another trigger.
The length of time the digitizer waits after it receives the trigger to start acquiring data.
Reference to the NI-SCOPE instrument session to pass to the next node in the program.
error out contains error information. This output provides standard error out functionality.
Where This Node Can Run:
Desktop OS: Windows
FPGA: Not supported
Web Server: Not supported in VIs that run in a web application