Adjusts the external calibration constants for an NI PXIe-4330 or NI PXIe-4331 device.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
reference voltage is the voltage across the AI+ and AI- terminals, measured using a DMM.
reference excitation is the voltage across the EX+ and EX- terminals, measured using a second DMM.
shunt location specifies the resistive leg of the Wheatstone bridge that has an external shunt resistor connected in parallel.
No shunt resistor connected.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application