Performs shunt calibration for the specified channels using a bridge sensor.
task in is the name of the task that the operation applies.
channels is a subset of virtual channels in the task that you want to calibrate. Use this input if you do not want to calibrate all the channels in the task, or if some channels in the task measure non-bridge-based sensors. If the input is empty, this node attempts to calibrate all virtual channels in the task.
shunt resistance specifies the shunt resistance in ohms.
shunt element location specifies the location of the shunt resistor.
Between Vch- and Vex+.
Between Vch- and Vex-.
Between Vch+ and Vex-.
Between Vch+ and Vex+.
shunt select specifies which shunt calibration switch to enable.
shunt source specifies whether to use an internal or external shunt.
Use the default shunt.
Use an internal shunt.
|External||Use an external shunt.|
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
bridge resistance specifies the bridge resistance in ohms. A value of -1 means to use the nominal bridge resistance specified when you created the virtual channel.
skip unsupported channels specifies whether or not to skip channels that do not support calibration. If skip unsupported channels is TRUE, this node calibrates only supported channels. If FALSE, this node calibrates the channels specified by channels. The default is FALSE.
task out is a reference to the task after this node runs.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application