Adjusts the external calibration constants for the timebase of a DSA device with an adjustable oscillator.
You must first call the DAQmx Setup DSA AO Timebase Calibration node and measure the frequency of the resulting sine wave. Repeat this process until calibration complete returns true.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
measured frequency specifies the measured frequency of the sine wave generated by the device.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
calibration complete returns TRUE if calibration of the device is complete. If this parameter returns FALSE, repeat the calibration process.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices