Converts a strain gauge voltage to units of strain.
The gauge factor of the strain gauge.
The voltage read from the strain gauge.
This input supports waveform and double-precision, floating-point data types.
The type of bridge configuration in which the strain gauge is mounted.
|Qtr Bridge I||0||Specifies a Quarter Bridge Type I configuration.|
|Qtr Bridge II||1||Specifies a Quarter Bridge Type II configuration.|
|Half Bridge I||2||Specifies a Half Bridge Type I configuration.|
|Half Bridge II||3||Specifies a Half Bridge Type II configuration.|
|Full Bridge I||4||Specifies a Full Bridge Type I configuration.|
|Full Bridge II||5||Specifies a Full Bridge Type II configuration.|
|Full Bridge III||6||Specifies a Full Bridge Type III configuration.|
Default: Half Bridge II
The excitation voltage you use.
Default: 3.33 V
The strain gauge nominal resistance value in ohms.
Default: 120 Ω
The unstrained voltage of the strain gauge after you mount it in its bridge configuration.
Read this voltage at the beginning of your application and save it to pass to this node.
The lead resistance. In many cases, the lead resistance is negligible and you can leave this terminal unwired.
This input is used when bridge configuration is set to quarter-bridge and half-bridge configurations. This input is ignored when bridge configuration is set to full-bridge configurations.
The return strain value.
This output returns a waveform or double-precision, floating-point data type depending on the data type of the Vsg input.
Where This Node Can Run:
Desktop OS: Windows
FPGA: This product does not support FPGA devices