If the status LED is blinking very quickly, a bad power supply may cause this behavior. The CompactRIO controller requires between 11 and 30 Volts of DC Power with sufficient current. Use a voltmeter to verify that your power source is outputting the proper level of voltage. If you find that you are not getting the proper amount of voltage or current out of the power supply, try switching to a different power supply.
A slower blinking - once every couple of seconds - means that the CompactRIO does not have an IP Configuration. Configuring the IP settings for the CompactRIO in the Measurement & Automation Explorer (MAX) should resolve this issue if the power supply is working properly. Please refer to the troubleshooting document in the related links section if you believe the power supply is working properly and are unsure of how to proceed configuring your device or are unable to configure it.
Note: More specific information about LED indications can be found in the Operating Instructions of your CompactRIO.
After checking the power supply voltage and/or changing the power supply, if you still experience the continuous flashing, please contact National Instruments Technical Support by going to ni.com/support.