This file contains important information about Calibration Executive, including installation instructions, new features, and known issues.
Calibration Executive provides product calibration support on Windows 32-bit and 64-bit operating systems. Note that Windows 32-bit and 64-bit support is dependent upon device driver support. Refer to the Windows Support topic in the Calibration Executive Help for information about exceptions.
Calibration Executive 5.0 does not support legacy procedures that require the NI Traditional DAQ driver. Calibration support for these devices is available using Calibration Executive for Traditional DAQ Version 3.5.2.
Calibration Executive has the following requirements:
Calibration Executive supports the following operating systems (64-bit operating system is recommended):
Note In 2016 National Instruments dropped support for Windows Vista, Windows XP, Windows Server 2003, and installations of Windows 7 without any service packs. Calibration Executive will not install or run on an unsupported OS. You cannot deploy or distribute applications that use Calibration Executive to an unsupported OS. Additionally, after installing this version of Calibration Executive, you cannot use any installers built on this computer with any version of LabVIEW, LabWindows™/CVI™, NI TestStand™, or Measurement Studio on an unsupported OS.
For more information about the changes to our OS support for 2016, refer to KB 79UC78LS, Why Does my LabVIEW, LabWindows/CVI, Measurement Studio, or TestStand Built Installer Fail on Windows XP/Vista and Server 2003?.
Note Support for Windows 32-bit operating systems may require disabling physical address extension (PAE). To learn how this might affect your system and what actions you might need to take, visit ni.com/info and enter the Info Code PAESupport.
1 NI software installs VC2015 Runtime and .NET 4.6.2. Windows 8.1 requires Microsoft updates to support these items. Refer to Microsoft KB2919442 and KB2919355 for more information about how to install these updates.
2 NI software is signed with a SHA-256 certificate. Windows 7 SP1 requires Microsoft updates to support SHA-256. Refer to Microsoft KB3033929 for more information about how to install this security update.
Ensure that your system meets the prerequisites listed in the System Requirements and Supported Operating Systems sections. In addition, all instrument drivers must be installed prior to installing Calibration Executive; see Driver Information for more information.
Before you begin, log onto your system as an administrator or as a user with administrator privileges. Be sure to exit all National Instruments applications.
To install Calibration Executive, follow these steps:
Note You must save the installer to your local drive. Do not attempt to open the file from its current location or to run the file from a temporary location.
Note If you are unable to locate your activated serial number for Calibration Executive, check the following file on your system: C:\ProgramData\National Instruments\Calibration Executive\CalExecLicenseInfo.ini.
Note To conserve disk space, you can delete the extracted files. It is recommended that you keep the extracted installation files in case you need to add features from this distribution in the future.
The following drivers are required for specific product support in Calibration Executive 5.0. Download drivers from http://www.ni.com/downloads/ and install manually.
The following instruments require specific drivers that are not installed by Calibration Executive. To install these drivers, you must download them directly from the manufacturer's website; you can find links to the third party instrument drivers at ni.com/idnet. The driver versions listed are those tested during development; later versions might be available and should be compatible.
Note This release does not include an NI-VNA driver. To run the PXI-5691 and PXIe-5632 procedures, download NI-VNA 16.0 and install the driver manually after you install Calibration Executive.
The Behavior Changes sections below list the procedures that have been modified since the Calibration Executive 4.0 major release, organized by release versions. Refer to an individual procedure's About Procedure section for details about the changes.
The Calibration Executive 5.0 release changes the following:
|PXIe-4480||PXIe-4481||PXIe-5654 (10 GHz)|
|PXIe-5654 (20 GHz)||PXIe-5840 (2 yr limits)||VB-8054|
|PXIe-5113||PXIe-5413 (1 CH)||PXIe-5413 (2 CH)||PCI-54213||PXI-54213|
|PXIe-5423 (1 CH)||PXIe-5423 (2 CH)||PXIe-5433 (1 CH)||PXIe-5433 (2 CH)||PXI-54413|
|PXIe-54424||PXIe-5646R||NI 9218 with DSUB||NI 9218 with LEMO|
The Calibration Executive 4.6.1 patch release changes the following:
The Calibration Executive 4.6 patch release changes the following:
|NI 9207||NI 9208||NI 9209||NI 9210 with mini-TC|
5 Supported PXIe-5668 models now include the 14 GHz variant (80 MHz, 200 MHz, and 765 MHz bandwidths) and the 26.5 GHz variant (80 MHz bandwidth).
|B/E/M/S/X Series Procedure (60xx/61xx/62xx/63xx)||PXIe-4154||PCI-5122|
|PXIe-5668 (26.5 GHz, 200 MHz & 765 MHz bandwidths)||PXIe-5840||PXIe-6738|
|PXIe-6739||NI 9207 (with DSUB)||NI 9212||NI 9361|
Note The cDAQ-9172 chassis is not supported in NI-DAQmx 17.6 and later. Therefore, cDAQ-9172 is no longer supported in Calibration Executive beginning with version 4.6. Refer to http://www.ni.com/product-documentation/54233/en/ for information about devices and modules no longer supported in NI-DAQmx.
The Calibration Executive 4.5.1 patch release changes the following:
|NI 9216||NI 9226||PXIe-5450||PXIe-5451|
The Calibration Executive 4.5 patch release changes the following:
|NI 9218||PXI-6683||PXI-6683H||PXIe-5185 (1 MΩ)|
|PXIe-5186 (1 MΩ)||PXIe-5413||PXIe-5423||PXIe-5433|
|PXIe-5160||PXIe-5162||PXIe-5185 (50 Ω)||PXIe-5186 (50 Ω)||PXIe-5442|
The Calibration Executive 4.2.1 patch release changes the following:
The Calibration Executive 4.2 minor release changed the following:
|NI 9250||NI 9251|
|NI 5650||NI 5651||NI 5652||PXIe-5663/E||PXIe-5673E|
The Calibration Executive 4.1.1 patch release changes the following:
The Calibration Executive 4.1 minor release changed the following:
|PXIe-4080 (90 day)||PXIe-4081 (90 day)||PXIe-4082 (90 day)|
|NI 9246||NI 9247||VB-8034|
|PXIe-4080 (2 yr)||PXIe-4081 (2 yr)||PXIe-4082 (2 yr)||PXIe-4138||PXIe-4139|
|RM-4339||PXIe-4464||PXIe-5160 (2 CH)||PXIe-5160 (4 CH)||PXIe-5162 (2 CH)|
|PXIe-5162 (4 CH)||PXIe-5673||PXIe-5673E (3.3 GHz)||PXIe-5673E (6.6 GHz)||PXIe-5644R|
|PXIe-5645R||PXIe-5646R||PXIe-5665 (3.6 GHz)||PXIe-5665 (14 GHz)||PXIe-6570|
The Calibration Executive 4.0.2 patch release changed the following:
The Calibration Executive 4.0.1 patch release changed the following:
|PXIe-4136||PXIe-4137||PXIe-4138||PXIe-4139||PXIe-5162 (4 CH)|
|NI VB-8012||NI 9207 with DSUB||NI 9208 with DSUB||NI 9209 with DSUB|
Note The Verify & Adjust Run Mode option was removed from the PXIe-5645R procedure in Calibration Executive 4.0 due to Windows system memory limitations. Instead, run Verify Only, Adjust Only, and Verify Only, and then merge reports to meet any Verify, Adjust, Re-verify requirements.
This section describes known issues with Calibration Executive.
|643618||Report generator can crash on large files in PDF format.|
|665119||Report generator can crash when compiling a large file in Microsoft Word and Excel formats.|
|677002||Some procedures require a PXI-4070, PXI-4071, or PXI-4072 DMM as an instrument standard; running these using a PXIe-4080, PXIe-4081, or PXIe-4082 DMM causes errors. For more information and a list of procedures, refer to the Calibration Executive Help topic DMM Instrument Standard Considerations.|
|677937||PXIe-5840 self-calibration runs out of memory on 32-bit operating systems. Refer to the Self-Calibration Memory Issue on 32-bit Operating System section in the PXIe-5840 procedure help for further details.|
Refer to the Calibration Executive Help, accessible from Help»Calibration Executive Help, for information about Calibration Executive.
You can automate the installation of most NI products using command-line arguments to suppress some or all of the installer user interface and dialog boxes. However, starting with the August 2012 releases (products using NI Installers version 3.1 or later), you may need to perform additional steps before or during a silent installation of NI software.
If the NI product you are installing uses Microsoft .NET 4.0, the .NET installer may run before any NI software installs and may require a reboot before the installation of NI software begins. To avoid a .NET reboot, install .NET 4.0 separately before you install NI software.
For more information about automating the installation of NI products, refer to the following KnowledgeBase articles:
Microsoft Windows 10 is the latest version of the Windows operating system and features significant changes compared to previous versions. Windows 10 introduces several new capabilities and also combines features from both Windows 7 and Windows 8. For more information about NI support for Windows 10, visit ni.com/windows10.
When you install NI software on Microsoft Windows 8.1, you will notice a few additional tiles in the Apps view, including shortcuts to NI application software products such as NI LabVIEW, Measurement & Automation Explorer (NI MAX), and NI Launcher. For more information about NI support for Windows 8.1, visit ni.com/windows8.
© 2005–2019 National Instruments. All rights reserved.
Under the copyright laws, this publication may not be reproduced or transmitted in any form, electronic or mechanical, including photocopying, recording, storing in an information retrieval system, or translating, in whole or in part, without the prior written consent of National Instruments Corporation.
NI respects the intellectual property of others, and we ask our users to do the same. NI software is protected by copyright and other intellectual property laws. Where NI software may be used to reproduce software or other materials belonging to others, you may use NI software only to reproduce materials that you may reproduce in accordance with the terms of any applicable license or other legal restriction.
End-User License Agreements and Third-Party Legal Notices
You can find end-user license agreements (EULAs) and third-party legal notices in the following locations after installation:
U.S. Government Restricted Rights
If you are an agency, department, or other entity of the United States Government ("Government"), the use, duplication, reproduction, release, modification, disclosure or transfer of the technical data included in this manual is governed by the Restricted Rights provisions under Federal Acquisition Regulation 52.227-14 for civilian agencies and Defense Federal Acquisition Regulation Supplement Section 252.227-7014 and 252.227-7015 for military agencies.
IVI Foundation Copyright Notice
Content from the IVI specifications reproduced with permission from the IVI Foundation.
The IVI Foundation and its member companies make no warranty of any kind with regard to this material, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose. The IVI Foundation and its member companies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material.
Refer to the NI Trademarks and Logo Guidelines at ni.com/trademarks for information on NI trademarks. Other product and company names mentioned herein are trademarks or trade names of their respective companies.
For patents covering the NI products/technology, refer to the appropriate location: Help»Patents in your software, the patents.txt file on your media, or the NI Patent Notice at ni.com/patents.