Refer to the following when making connections to the back panel of the NI-4134 and to understand the behavior of the back panel LED indicators.
|1||TRIG OUT 0||SMA (female)||Sends a trigger to external equipment||External equipment trigger input; tighten to maximum torque of 90 N⋅cm (8 lb⋅in)|
|2||TRIG OUT 1||SMA (female)|
|3||TRIG IN||SMA (female)||Receives a trigger from external equipment||Trigger source; tighten to maximum torque of 90 N⋅cm (8 lb⋅in)|
|4||HIGH SPEED PULSE SECONDARY||BNC (female)||High speed pulse signal output; also passes SMU Sense HI|
|5||HIGH SPEED PULSE PRIMARY||BNC (female)||High speed pulse signal output; also passes SMU Force HI||DUT load board|
|6||SCOPE OUT||BNC (female)||100× attenuation output of signal from HIGH SPEED PULSE PRIMARY to allow an oscilloscope to measure the pulse signal; 50 Ω source termination||50 Ω oscilloscope input channel|