HD BNC connector for enabling noise source on or off to facilitate noise figure measurements.
Connects the analog RF input signal to the PXIe-5698. The maximum RF input is +17 dBm.
Outputs the bypassed or preamplified RF signal. The maximum RF output is +25 dBm.
Indicates the basic hardware status of the module.
OFF—The module is not yet functional, or the module has detected a problem with a PXI Express power supply rail.
AMBER—The module is being accessed. Accessed means that the device setup registers are being written to in order to control the device.
GREEN—The module is ready to be programmed by NI-RFSA.
Indicates the module state.
OFF— The module is in a quiescent state.
AMBER—The module is waiting for a Reconfiguration Trigger from the configuration list.
GREEN—The module is triggered and is running a step from the configuration list.
RED—The module has detected an error state. An error state may indicate the module has exceeded approved operating temperature and thermal shutdown has occurred or that the module has detected a power supply failure. If the power supply fails, contact NI technical support.