Associates TEDS information with the physical channel you specify.
If you do not specify the filename of a data sheet in the virtual TEDS file input, this node attempts to find a TEDS sensor connected to the physical channel. This node temporarily overrides any TEDS configuration for the physical channel that you performed in MAX.
physChan in is the name of the physical channel you want to configure. A DAQmx physical channel constant, or control, lists all the physical channels, on devices and modules installed in the system.
You also can wire a string that contains a physical channel name to this input.
virtual TEDS file is the path to a Virtual TEDS data sheet that you want to associate with the physical channel. If you do not wire anything to this input, this node attempts to find a TEDS sensor connected to the physical channel.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
physChan out is the name of the physical channel the operation applied to. A DAQmx physical channel constant, or control, lists all the physical channels, on devices and modules installed in the system.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application