Adjusts the internal and external calibration constants for an NI 4204 device. You must connect a known voltage to the device and specify that voltage with reference voltage.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
physical channels specifies the physical channel(s) to calibrate.
lowpass filter cutoff frequency specifies in Hz which lowpass filter cutoff frequency to calibrate.
sample and hold enable specifies whether to calibrate the channel(s) with the sample and hold circuitry of the channel enabled.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
reference voltage specifies in volts the known voltage to use as a reference for calibration.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application