Adjusts the external calibration constants for an NI 9264 device.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
physical channels specifies the names of the physical channels to use to create virtual channels. The DAQmx physical channel constant lists all physical channels on devices and modules installed in the system.
You can also wire a string that contains a list, or range of physical channels, to this input. If you have an array of physical channels, use the DAQmx Flatten Channel String node to convert the array to a list.
reference value specifies the reference value measured using a calibrator.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application