Adjusts the external calibration constants for the analog input section of a DSA device.
You must connect a known voltage to the device and specify that voltage with reference voltage.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
reference voltage specifies in volts the known voltage to use as a reference for calibration.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
inputs shorted specifies whether the input terminals are shorted for the calibration procedure. Refer to the calibration procedure for your device for more information.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application