You must configure and rename your instruments in Measurement & Automation
Explorer (MAX) before you begin taking measurements.
(Chamber revB) Before you begin, verify that the LED status
indicator on the chamber exterior is solid green.
In the configuration tree, expand Devices and Interfaces
to see the list of installed NI hardware. Installed instruments appear under the
name of the associated chassis.
Within the Settings section for each instrument, enter the
new name in the Name textbox as shown in the following
|PXIe-6361 (p/n 781055-01)
Click Save in the MAX toolbar after you change the name
for each instrument.
In the configuration tree, select the PXIe-1095 chassis.
On the Triggers tab, route the following PXI trigger lines
between the PXIe-6361 (p/n 781055-01) and PXIe-5831:
Set PXI_Trig0 to be away from the PXIe-6361 (p/n 781055-01) bus segment to the PXIe-3622 bus segment.
Set PXI_Trig1 to be away from the PXIe-3622 bus segment to the PXIe-6361 (p/n 781055-01) bus segment.