You can perform averaged measurements on an FFT channel to improve measurement accuracy or to help compensate for a low signal-to-noise
Disabled—Disables FFT averaging.
RMS—Root mean square. RMS averaging averages the energy (or power) of the signal, so it reduces signal fluctuations, but not
the noise floor.
Peakhold—Performs a peak hold averaging measurement at each individual frequency line and retains the RMS peak levels of the averaged
quantities from one FFT record to the next. Peak-hold averaging is the most useful when configuring a measurement system or
when applying limit or upper limit testing to a frequency spectrum.