You can add your digital lines to the project by configuring signals in the Channel
Specification. Configure digital signals by enabling the line and determining the input or
Select one or more channels you wish to configure.
Hover over a channel row and click the Configure gear
The configure gear appears when selecting multiple channels of the same data
type (analog, digital, counter, calculated, and so on). If channels of
different data types are selected, the configure gear will not appear.
Specify the channel name in the Name field.
Digital from the
Physical Measurement pull-down selector.
Available channel configuration options vary for digital input or digital output. For digital input lines,
Line will automatically populate the
Sensor option. For digital output, you can select the output level.
For digital input, record any sensor information or test procedures in the
Sensor Properties field, if applicable. To edit the
Sensor Properties field, complete the following
The information will populate the Sensor
Properties field of the channel configuration.
Click the Sensor Properties edit icon at the top
right of the box.
Fill in the information in the Sensor Properties
dialog that opens.
For digital output, specify the output Value for the
channel or map to an input channel Source to use as the
channel output, if applicable.
FlexLogger does not support waveform
output. Output signals are software timed and nondeterministic.
Apply any additional applicable configuration options to the line. Refer to the
Configuring I/O Channels topic for more information on available
For digital input, after the
Physical Measurement is selected,
Live value shows the value of the signal being measured by the line.
FlexLogger does not log output channels.