You can add your counters to the project by configuring signals in the Channel Specification. You configure counters by enabling the line and determining the settings.
Select one or more channels you wish to configure.
Hover over a channel row to see the
. Click the gear for the channel or channels you wish to configure.
The configure gear appears when selecting multiple channels of the same data type (analog, digital, counter, calculated, and so on). If channels of different data types are selected, the configure gear will not be available.
Specify the channel name in the
field, if necessary.
you wish to acquire for this channel.
For information on which physical counter measurement types are available for the sensor class you are using, refer to the
Supported Sensor Class Types
that will be used to take the counter measurement.
options that appear are specific to the selected sensor, if applicable.
If applicable, record any sensor information or test procedures in the
field. To edit the
field, complete the following steps:
The information will populate the
field of the channel configuration.
edit icon at the top right of the box.
Fill in the information in the
dialog that opens.
If applicable, set the configuration options to the desired settings for the corresponding counter connected to the channel.
When selecting multiple channels, available configuration options may not be available for channels with different requirements and specifications. Available channel configuration options vary by counter measurement type.
If necessary, apply any scaling options to the counter. Refer to the
Scaling Electrical Values to Physical Values
topic for more information on available options.
After a Sensor Class is selected,
shows the raw electrical value being measured from your sensor without any scaling applied to it.
shows the sensor value with any scaling configurations applied. Raw values are meant for validation of the sensor configuration and are not logged to a data file.