Use the Calibrate channels feature to apply calibration to multiple channels
In your Channel Specification document, select all configured channels you wish
In your Channel Specification document toolbar, click the Calibrate
channels button .
Select one of the following calibration options:
- Select Zero to apply an offset to the analog input channels to
obtain a zero reading. Zero calibration is only available for Linear and
Sensitivity scaling types.
- Select Null offset to perform offset nulling calibration on the
strain channels to adjust the output of the bridge to 0 V.
- Select Shunt calibrate to adjust for errors on your bridge
circuits caused by resistance of the lead wires between the EX pins on the
device and the strain gage. For more information, refer to the Shunt Calibrating One or More Channels
FlexLogger automatically applies the selected offset to all
selected channels that support that calibration type.
Multiple channel calibration
cannot be applied when the test is running.