You can change or configure the interface (port) settings from within software using
the Interface Settings dialog, if necessary.
LIN interfaces are software representations of physical ports. Complete the following steps to adjust the port settings.
Termination, if necessary.
Termination schemes might result in bus error frames or missed data. Refer to the LIN Cable Termination topic for more information on proper termination.
Termination settings can only be configured or changed when the test is
Baud rate pull-down menu to select an applicable baud rate to all clusters, if necessary.
The port will use the baud rate defined by the selected database. A
Baud rate set to an unsupported value might result in bus error frames or missed data.
(Optional) After the
Baud rate is selected, the option to
Enable raw frame logging is enabled. Selecting this option will allow FlexLogger to log data without associating the port to a database.
Using the Database drop-down menu, select the database
file that contains your signal definitions.
Using the Cluster drop-down menu, select the desired
cluster from your database.
All available frames and the signals they contain display in the
Using the LIN schedule drop-down menu, select the
desired LIN schedule.
The following considerations apply to LIN
- Assigning a schedule to a LIN port automatically specifies that port as
- LIN ports will not show data if a master port is not specified.
- LIN schedules can only be configured or changed when the test is not
Schedule information will be displayed in the details column of the
corresponding signal in the Channel