The SLSC-12251 and SLSC-12252 modules provide current sensing on each DUT channel. The
ECUTS includes additional
instrumentation necessary to perform current measurements.
To read the current from SLSC channels
using the TestStand ECU Software Toolkit, complete the following steps.
Open or create a sequence file in TestStand.
Insert an SLSC FIU – Read Current step from
IO Configuration»Instrument Control »SLSC 12251/2 in
the Insertion Palette pane.
Configure the step in the Step Settings pane by selecting the Switch Executive
Virtual Device associated with your SLSC module.
Click the Switch Executive
icon to open the selected Switch Executive Virtual Device in Switch
Click the + button to add current measurement
Assign a numeric variable for each channel to which the
data will be saved.
Use the Limits tab to configure a comparison type for the step to pass/fail the
step according to the acquired measurement.