The ECUTS does not ship ready for use with
facility compressed air. To configure the ECUTS
pneumatics to route compressed air, use the following guidelines:
- Use 6 mm outer diameter tubing to connect to the pneumatic air port on the back of
the test rack.
- Mount pneumatic components such as regulators and valves on the bottom of the test
or in empty test rack space.
- If installed, do not remove the ECUTS
ballast plates from the bottom of the test rack. Removing the ballast plates may
cause the ECUTS to become unstable.
- Connect to the pneumatic output ports using 6 mm outer diameter
Available pneumatic output ports on the mass interconnect are J9,
J10, or J16 (if populated), in the pin 1 location.
The mass interconnect receiver has a contact switch that closes when an ITA mates to the
receiver. You can use the contact switch to control a valve to stop the flow of air when
removing a fixture from the mass interconnect.
The mating connection on the ITA is a VPC Pneumatic Contact 610132106, 610132110,
610132111, or 610132112 in the VPC Mini Power Module 510108115.