Install the application software (if applicable), as described in the installation instructions that accompany your software. Check your driver and application development environment (ADE) readme files for specific version compatibility.
Carefully align and connect one end of the Ethernet cable to Ethernet port 0 on the device, and the other end directly to your computer or any network connection on the same subnet as your computer.
Power the device by carefully aligning and connecting the power IN port on FieldDAQ to an external 9 V DC to 30 V DC power source with the 5-pinL-coded M12 power cable.
The Power LED lights.
To add the FieldDAQ to your software configuration, open NI MAX on your Windows host computer. Expand Devices and Interfaces»Network Devices.
If the device is on your local subnet, the device automatically appears in the list of available devices. Right-click the FieldDAQ device and select Add Device.
If the device is not on your local subnet, right-click Network Devices and select Find Network NI-DAQmx Devices.
In the Find Network NI-DAQmx Devices dialog box that opens, do one of the following:
Check the box that corresponds to your device in the Hostname column.
Figure 1. Find Network NI-DAQmx Devices
If you know the device IP address, such as 192.168.0.2, enter it into the Add Device Manually field of the Find Network NI-DAQmx Devices window, and click the + button.
Enter the hostname of the device. The default hostname is FD116xx-<serial number>.
Click Add Selected Devices.
The FieldDAQ device icon changes from white to dark grey, indicating that it is recognized and present on the network.
Figure 2. MAX Icons and States
Discovered, but Not Added to the Network
Recognized, Present, and Reserved on the Network
If your device does not appear in Available Devices, click Refresh List. If the device still does not appear, try the following:
If you connected the FieldDAQ device directly to your computer, ensure your network card is configured to obtain an IP address automatically, then click Refresh List.
If you connected the FieldDAQ device directly to your computer, the setup time may be longer. Wait 30 to 60 seconds after the STATUS LED turns off, then click Refresh List.
Contact your system administrator to confirm that the network is working and that a firewall is not interfering with discovery. For additional troubleshooting resources for the FieldDAQ device, refer to the Troubleshooting Device Connectivity in this guide and the Finding a Network DAQ Device in MAX topic in the Measurement & Automation Explorer Help for NI-DAQmx.
Figure 3. Network Devices in MAX
If the FieldDAQ device is not reserved automatically, select the device and click the Reserve Network Device button.
Self-test your device in MAX by expanding Devices and Interfaces»Network Devices, right-clicking your FieldDAQ device, and selecting Self-Test. Self-test performs a brief test to determine successful device installation. When the self-test finishes, a message indicates successful verification or if an error occurred. If an error occurs, refer to ni.com/support/daqmx.
Run a Test Panel in MAX by expanding Devices and Interfaces»Network Devices»your FieldDAQ device, right-clicking the bank of connectors in your FieldDAQ device, and selecting Test Panels. If the test panel displays an error message, refer to ni.com/support. Click Close to exit the test panel.
When in use, the FieldDAQ device may become warm to the touch. This is normal.
M12 connectors must be mated to cables or have caps installed on them to meet IP requirements. Cover the unused connectors with the included plastic caps whenever water, dust, or dirt are present.
Avoid long periods of exposure to sunlight.
Recently Viewed Topics