The NI-5782 allows you to configure the clock, ADC, sampling, and other settings with the User Command, User Data 0, and User Data 1 signals.
|User Setting||User Command||User Data 0 Value||User Data 1 Value||User Return Value|
|Clock Settings||0||0 = Internal Sample Clock
1 = Internal Sample Clock locked to an external Reference Clock through the CLK IN connector
2 = External Sample Clock through the CLK IN connector (disables the CLK OUT connector)
3 = Internal Sample Clock locked to an external Reference Clock through Sync Clock
4 = External Sample Clock through the Sync Clock
|Default: 0 = 500 MHz DAC output clock, 1 GHz timebase, no interpolation.
Refer to the User Data Clock Settings table for more information.
|Read User EEPROM||6||EEPROM address to read data.||N/A||Read data.|
|Write User EEPROM||7||EEPROM address to write data.||Write data.||N/A|