The PXIe-4163 includes a self-calibration feature that you can use to significantly reduce many error sources, such as voltage source and measure offset error, voltage source and measure gain error, and current source and measure offset error.
When you run self-calibration, the output terminal is disconnected. Low-amplitude, low-energy glitches may appear at the output, but in most circumstances, these glitches are not noticeable. Self-calibration on the PXIe-4163 is typically under 10 seconds.
When calling niDCPower Cal Self Calibrate (niDCPower_CalSelfCalibrate), specify all channels of your PXIe-4163 with the channel name input. You cannot self-calibrate a subset of PXIe-4163 channels.
For optimum accuracy, use self-calibration in the following situations:
The PXIe-4163 incorporates a temperature sensor that is used to determine when the temperature changes ±5 °C from the previous calibration. When the most recent self-calibration time and temperature are queried using niDCPower Get Self Cal Last Date And Time (niDCPower_GetSelfCalLastDateAndTime) or niDCPower Get Self Cal Last Temp (niDCPower_GetSelfCalLastTemp), the value returned is from the most recent self-calibration, regardless of the Self Calibration Persistence property (NIDCPOWER_ATTR_SELF_CALIBRATION_PERSISTENCE attribute). When the one-year calibration interval expires, an external calibration is required.