Specifies a hysteresis level in the units of the measurement. If Ref.AnlgEdge.Slope is Rising, the trigger does not deassert until the source signal passes below Ref.AnlgEdge.Lvl minus the hysteresis. If Ref.AnlgEdge.Slope is Falling, the trigger does not deassert until the source signal passes above Ref.AnlgEdge.Lvl plus the hysteresis. Hysteresis is always enabled. Set this property to a non-zero value to use hysteresis.
Long Name: Trigger:Reference:Analog Edge:Hysteresis
Class: DAQmx Task
Where This Property Is Available:
Desktop OS: Windows
Web Server: Not supported in VIs that run in a web application