Creates a channel that uses a special purpose counter to take a timestamp and synchronizes that counter to a GPS receiver. With the exception of devices that support multi-counter tasks, you can create only one counter input channel at a time with this node, because a task can contain only one counter input channel. To read from multiple counters simultaneously, use a separate task for each counter. Connect the input signals to the default input terminals of the counter, unless you select different input terminals.
task in specifies the task to which to add the virtual channels this node creates. If you do not specify a task, NI-DAQmx creates a task for you, and adds the virtual channels this node creates to that task.
counter specifies the name of the counter to use to create the virtual channel. The DAQmx physical channel constant lists all physical channels, including counters, for devices installed in the system.
name to assign specifies a name to assign to the virtual channel this node creates. If you do not wire a value to this input, NI-DAQmx uses the physical channel name as the virtual channel name.
If you use this input to provide your own names for the virtual channels, you must use the names when you refer to these channels in other NI-DAQmx nodes, such as the source input of the DAQmx Trigger node.
If you create multiple virtual channels with one DAQmx Create Virtual Channel node, you can specify a comma-separated list of names to assign to the virtual channels. If you provide fewer names than the number of virtual channels you create, NI-DAQmx automatically assigns names to the virtual channels.
units specifies the units to use to return the timestamp.
|From Custom Scale||
Units a custom scale specifies. If you select this value, you must wire a custom scale name to the custom scale name input.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
gps synchronization method specifies the method to use to synchronize the counter to a GPS receiver.
Use the IRIG-B synchronization method. The GPS receiver sends one synchronization pulse per second, as well as information about the number of days, hours, minutes, and seconds that elapsed since the beginning of the current year.
Do not synchronize the counter to a GPS receiver. The timestamp measurement returns the number of seconds that elapsed since the device powered up, unless you set the CI.Timestamp.InitialSeconds property.
Use the PPS synchronization method. The GPS receiver sends one synchronization pulse per second, but does not send any timing information. The timestamp measurement returns the number of seconds that elapsed since the device powered up, unless you set the CI.Timestamp.InitialSeconds property.
custom scale name specifies the name of a custom scale for the channel.
task out is a reference to the task after this node runs. The task contains any newly created virtual channels. If you did not wire a value to task in, NI-DAQmx automatically creates the task this output refers to.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices
Web Server: Not supported in VIs that run in a web application