Generates a specified voltage from the device.
You must measure the voltage and pass that value into the DAQmx Adjust 4463 Calibration node.
calHandle in is a reference to the calibration session that you created using the DAQmx Initialize External Calibration node.
physical channels specifies the physical channel(s) to calibrate.
terminal configuration specifies the terminal configuration of the device.
gain specifies the gain setting to calibrate, in decibels.
output voltage specifies, in Hz, the output voltage to generate.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
calHandle out is a reference to the calibration session. Wire this output to other external calibration nodes.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices