Configures the task to stop the acquisition when the device acquires all pretrigger samples; an analog signal enters or leaves a range you specify; and the device acquires all post-trigger samples.
task in is the name of the task that the operation applies.
source is the name of a virtual channel or terminal where there is an analog signal to use as the source of the trigger.
For E Series devices, if you use a virtual channel, it must be the only channel in the task. The only terminal you can use for E Series devices is PFI0.
window top is the upper limit of the window. Specify this value in the units of the measurement or generation.
window bottom is the lower limit of the window. Specify this value in the units of the measurement or generation.
error in describes error conditions that occur before this node runs. This input provides standard 'error in' functionality.
when specifies whether the Reference Trigger occurs when the signal enters the window or when it leaves the window. Use window bottom and window top to specify the limits of the window.
Trigger when the signal enters the window.
Trigger when the signal leaves the window.
pretrigger samples per channel specifies the minimum number of samples to acquire per channel before recognizing the Reference Trigger. The number of post-trigger samples per channel is equal to number of samples per channel in the DAQmx Timing node minus pretrigger samples per channel.
task out is a reference to the task after this node runs.
error out contains error information. If error in indicates that an error occurred before this node ran, error out contains the same error information. Otherwise, error out describes the error status that this node produces.
Where This Node Can Run:
Desktop OS: Windows
FPGA: LabVIEW NXG does not support FPGA devices