The NI 7975R can access eight trigger lines, named PXI_Trig<0..7>, on the PXI Express backplane. You can use these trigger lines to communicate between multiple devices in the same chassis.
You can also use the following two additional PXI Express triggers.
When developing a high-level VI on the NI 7975R that uses triggers, reserve the trigger lines you are using and, to ensure compatibility with other NI FlexRIO devices, configure trigger pulses to last for at least two clock cycles of the clock on the receiving device. For example, if the clock on the receiving device is 80 MHz, which is a clock period of 12.5 nanoseconds, the trigger line must be constant for at least 25 nanoseconds, which is two cycles of an 80 MHz clock.
The clocks between an NI FlexRIO PXI device and another PXI device might not be perfectly synchronized. If you assert a trigger line on an NI FlexRIO PXI device, you cannot determine at what point in the clock period the trigger registers in the receiving flip-flop. If the trigger arrives during the setup or hold time of the receiving flip-flop, you cannot determine the state of the line for that clock period. Asserting the trigger pulse for two clock cycles ensures that at least one clock cycle on the receiving flip-flop registers as a rising edge and transfers as a trigger.