The counter can output a single pulse. The pulse appears on the Counter n Internal Output signal of the counter.
You can specify a delay from when the counter is armed to the beginning of the pulse. The delay is measured in terms of a number of active edges of the Source input.
You can specify a pulse width. The pulse width is also measured in terms of a number of active edges of the Source input. You also can specify the active edge of the Source input (rising or falling).
The following figure shows a generation of a pulse with a pulse delay of four and a pulse width of three (using the rising edge of Source).